Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Published (Last):||4 April 2009|
|PDF File Size:||12.76 Mb|
|ePub File Size:||12.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
If you click on average icon, max. Press Ok for launching measurement. Average – Rec Recording Veloci. By pressing ikmo button you will print only current page.
KIMO MP 200 manometer
Select Start with access key for launching measurement. If the probe has been just connected, the screen indicates the remaining time of jimo up then the measurement screen appears, indicates the concentration and places it on a bargraph. Select compensation It is possible to modify the value of the compensation in temperature. Probe display Available information are: In regards of Quality insurance norms, we recommend that the instruments are checked once a year. Configuration If you use thermocouple probes, you must enter type into the Configuration sub-function.
Preview of tables of points of datasets Dataset summary You can display tables of points of datasets performed on the device. Summary screen of selected dataset is displayed From this screen, you can: The numbers of points is displayed.
When the measurement is fixed, press Min Max to access to maximum and minimum values of the measurement. Select other dataset using arrow keys t and u. To modify the period, press Period with the access key.
Kimo MP Multi Function Manometers | Pullman Instruments
The ”Probe” menu only appears when probes or module are connected. Launch a planned dataset A planned dataset is composed of several locations. This function can also be automatically performed by using the solenoid valve function. You must enter duration: Got it, continue to print. Press 1x in order to select HOLD function: Kio to Recording menu. Go to Recording menu.
The number of points selected and the parameter are displayed. Don’t show me this message again. Table Of Contents actoolsupply.
Preview of tables of points of datasets Dataset summary You can ,p200 tables of points of datasets performed on the device. Remove the front part at the back of the instrument. You can choose between a planned or a continuous dataset. You must enter duration: I – Technical Specifications actoolsupply.
Kkmo can access this function selecting COmax with the access key CO max The CO is measured on an adjustable period, the maximum value measured in this period is called CO max. For each location, the operator can enter a theorical value and a tolerance for the parameter to be controlled. This manual also for: The operator can choose an automatic or a manual dataset, with an instant value or an average.
The beep repetition frequency will increase accordong to the increase of detected gas concentration, and the display will blink at the same frequency. When the solenoid valve is enabled, it runs every minute. Click on Esc to return to previous screen. Air velocity with Pitot tube: This datasets can’t be set using Datalogger Software. Press 2x in order to select Min-Max function: Numbers of selected points and parameter for which calculation is carried out, are displayed For adding a new measuring point to this calculation, press OK to confirm.
You can choose between a planned or a continuous dataset. Use the arrows to modifiy the threshold value then press Validate to validate alarm threshold and back to measurement screen. For measurement launching, click on OK with the access key. Table of Contents Add to my manuals Add.
If Other is selected, you must enter a value. This menu allows kio view probe information plugged to C2, Module, C1 or wireless connections. If the probe is connected for more than 1 minute, measurement screen appears and indicates the concentration and places it on a bargraph. You can view each measuring points if you click on Visu.
MP Portable Thermo-Anemo-Manometer
On disabled mode, the probe is connected, the measurement is not carried out and the value is not displayed. Maintenance V — General informations Maintenance KIMO performs calibration, adjustment and maintenance of all your instruments to guarantee a constant level of quality of your measurements. Air velocity with Pitot tube: From this screen, press Measure to back to measurement.